Study: MP3 Player Headphones May Interfere With Pacemakers
November 10, 2008 3:22 AM
comment(s) - last by
Research indicates the MP3 player headphones, not the player itself, poses a health risk to some owners
If you listen to an MP3 player and have a heart pacemaker, you could be at risk of complications, medical researchers warn.
But unlike previously thought, the MP3 player is not the culprit -- the headphones plugged into the MP3 player pose a risk. Earlier in 2008, the U.S. Food and Drug Administration (FDA) officially concluded that MP3 players are safe to use regardless of their location.
Research presented during the American Heart Association conference indicates that some headphones commonly used by MP3 player owners can interfere with pacemakers or implanted defibrillators -- even if they're not being used and are disconnected.
Researchers led by Dr. William Maisel at the Medical Device Safety Institute at Beth Israel Medical Center in Boston
tested eight different models of headphones
and ear buds that were connected to Apple iPod MP3 players. During the study with 60 patients, 30 percent of subjects with defibrillators and 15 percent of subjects with pacemakers suffered interference due to their headphones.
"For patients with pacemakers, exposure to the headphones can force the device to deliver signals to the heart, causing it to beat without regard to the patients' underlying heart rhythm," Dr. Maisel said in a press statement. "Exposure of a defibrillator to the headphones can temporarily deactivate the defibrillator."
A strong magnetic substance inside some popular MP3 headphones mistakenly interrupt a pacemaker's beat or deactivate the defibrillator.
If placed around one inch from a device, there was interference one-fourth of the time -- in 10 patients out of 33 who had defibrillators and four of 27 overall patients with pacemakers. One pacemaker unexpectedly reset itself during the trials.
Sixty patients took part in the study.
Dr. Maisel said people shouldn't "overreact" to the research, but still advised patients with pacemakers or defibrillators to keep headphones at least 3 centimeters away from the implanted devices. Furthermore, headphones should not be placed in a breast pocket or dangle over their chest.
This article is over a month old, voting and posting comments is disabled
11/10/2008 5:54:33 AM
Most of music around today is so bad that even without a pace maker people can have a heart attack.
"We shipped it on Saturday. Then on Sunday, we rested." -- Steve Jobs on the iPad launch
Former Android OS Chief Heading New Google Robotics Group
December 4, 2013, 1:20 PM
Dow Chemical to NYC City Council: You Don't Even Know What Styrofoam is!
December 2, 2013, 8:30 PM
Retired Israel Air Force Commander: UAVs Need to Dogfight Before Replacing Fighter Pilots
November 27, 2013, 3:59 PM
Eating High-Fat Diet During Puberty May Increase Speed of Breast Cancer Development
November 27, 2013, 2:15 PM
Mayor Bloomberg Says No More Polystyrene Foam Dishes in NYC, Says it Can Cause Cancer
November 25, 2013, 11:51 AM
NASA Rejects Billionaire's Mission to Mars
November 22, 2013, 12:31 PM
Most Popular Articles
NSA Snares Americans' Porn Viewing Histories in Effort to Target Muslims
December 1, 2013, 9:00 PM
Coalition of 20+ Tech Firms Backs MRAM as Potential DRAM, NAND Replacement
November 29, 2013, 11:59 PM
Fed Up With Cheating OEMs, Microsoft Trolls Chromebooks in New Ad
November 27, 2013, 4:09 PM
Xbox? PCs? Mobile? Microsoft Wants One Windows to Rule Them All
November 25, 2013, 8:21 PM
Seattle Restaurant Bans Google Glass, Tells Wearers to "Just shut up and get out"
November 27, 2013, 10:27 AM
Latest Blog Posts
Global Cyber Espionage Concerns Reveal Growing Cyber Armies
Nov 29, 2013, 11:04 AM
Is The Period Becoming an Expression of Anger?
Nov 26, 2013, 2:02 PM
NSA and Congress -- You Will Never Kill the Constitution, It's an Idea
Nov 10, 2013, 2:00 PM
AT&T Explores $100B+ USD Deal to Acquire Vodafone's European Operations
Nov 4, 2013, 7:34 AM
U.S. Army Developing Cyber, Electronic War Arsenal
Oct 31, 2013, 4:49 PM
More Blog Posts
Copyright 2013 DailyTech LLC. -
Terms, Conditions & Privacy Information