Study: MP3 Player Headphones May Interfere With Pacemakers
November 10, 2008 3:22 AM
comment(s) - last by
Research indicates the MP3 player headphones, not the player itself, poses a health risk to some owners
If you listen to an MP3 player and have a heart pacemaker, you could be at risk of complications, medical researchers warn.
But unlike previously thought, the MP3 player is not the culprit -- the headphones plugged into the MP3 player pose a risk. Earlier in 2008, the U.S. Food and Drug Administration (FDA) officially concluded that MP3 players are safe to use regardless of their location.
Research presented during the American Heart Association conference indicates that some headphones commonly used by MP3 player owners can interfere with pacemakers or implanted defibrillators -- even if they're not being used and are disconnected.
Researchers led by Dr. William Maisel at the Medical Device Safety Institute at Beth Israel Medical Center in Boston
tested eight different models of headphones
and ear buds that were connected to Apple iPod MP3 players. During the study with 60 patients, 30 percent of subjects with defibrillators and 15 percent of subjects with pacemakers suffered interference due to their headphones.
"For patients with pacemakers, exposure to the headphones can force the device to deliver signals to the heart, causing it to beat without regard to the patients' underlying heart rhythm," Dr. Maisel said in a press statement. "Exposure of a defibrillator to the headphones can temporarily deactivate the defibrillator."
A strong magnetic substance inside some popular MP3 headphones mistakenly interrupt a pacemaker's beat or deactivate the defibrillator.
If placed around one inch from a device, there was interference one-fourth of the time -- in 10 patients out of 33 who had defibrillators and four of 27 overall patients with pacemakers. One pacemaker unexpectedly reset itself during the trials.
Sixty patients took part in the study.
Dr. Maisel said people shouldn't "overreact" to the research, but still advised patients with pacemakers or defibrillators to keep headphones at least 3 centimeters away from the implanted devices. Furthermore, headphones should not be placed in a breast pocket or dangle over their chest.
This article is over a month old, voting and posting comments is disabled
11/10/2008 5:54:33 AM
Most of music around today is so bad that even without a pace maker people can have a heart attack.
"We can't expect users to use common sense. That would eliminate the need for all sorts of legislation, committees, oversight and lawyers." -- Christopher Jennings
Obama Admin Looks to Revive Bill for Greater Media Protection in Wake of AP Controversy
May 16, 2013, 1:47 PM
Notre Dame Designs Camera to Detect Age of Bruise for Abuse Checks
May 16, 2013, 9:05 AM
Report: Climate Change Could Wash Away Historic Jamestown Site Within 100 Years
May 15, 2013, 5:39 PM
U.S. Navy Launches First Unmanned X-47B Aircraft from Carrier Flight Deck
May 15, 2013, 12:13 PM
Government Creates Global Warming "Refugee" Crisis in Alaska
May 15, 2013, 6:37 AM
Quick Note: ISS Astronaut Records David Bowie Cover from Space
May 13, 2013, 1:01 PM
Most Popular Articles
Report: Microsoft Eyes Return to "Dying" Windows 7 Path After Windows 8 Flop
May 13, 2013, 9:50 AM
Bill Gates Gets Teary-Eyed While Discussing Steve Jobs, Shows Off Life-Saving Tech on 60 Minutes
May 13, 2013, 12:30 PM
Windows 8.1 Will Be Free; Microsoft Holds Onto Struggling ARM Variant
May 14, 2013, 2:57 PM
Google Announces "Pure" Galaxy Nexus S4 for $649, Android Updates
May 15, 2013, 1:42 PM
U.S. Federal Traffic Board Wants to Make Drunk Driving Threshold Far Harsher
May 15, 2013, 11:32 AM
Latest Blog Posts
Parents of Pre-Teen Drivers Commonly Practice Distracted Driving Says Study
May 9, 2013, 7:16 AM
Apple's iOS 7 Running Into Internal Delays Due to Massive Overhaul
May 1, 2013, 4:26 PM
Elon Musk Willing to Spend More Money on Widening of 405 Freeway
Apr 26, 2013, 7:28 PM
New $100 Bill Due in October
Apr 24, 2013, 11:52 PM
Volkswagen Shows off iBeetle in Shanghai
Apr 22, 2013, 9:29 PM
More Blog Posts
Copyright 2013 DailyTech LLC. -
Terms, Conditions & Privacy Information